Fluke Corporation is voluntarily recalling certain Digital Clamp Meters that were manufactured between Sept. 1, 2010 and Oct. 31, 2012.
Name of products: Certain Fluke 373, 374, 375, and 376 Digital Clamp Meters
Manufacturer: Fluke Corporation
Hazard: The printed circuit assembly in these units may not be properly fastened to the test lead input jack. This may result in inaccurate voltage readings, including a low or no voltage reading on a circuit energized with a hazardous voltage, presenting a shock, electrocution, or thermal burn hazard.
Description: This voluntary recall involves Fluke Digital Clamp Meters with the model and serial numbers listed below. Any serial number preceded or followed by the letter “R” is NOT affected by this recall notice.
|Model||Manufacture Dates||Starting Serial Number||Ending Serial Number|
|Fluke 376||Sept 2010 Feb 2011||14270001||15909999|
|Fluke 375||Sept 2010 Mar 2011||14270001||16079999|
|Fluke 374||Oct 2010 Feb 2011||14270001||16379999|
|Fluke 373||Oct 2010 Oct 2012||14270001||21950000|
If you own one of these clamp meters, please stop using it immediately and send it back to Fluke for repair. If someone else within your organization is the primary user of the Fluke 37X Series Clamp Meter or is responsible for responding to recall notices, please pass this notice along to the appropriate person. (Please send only the clamp meter; any test leads/boxes/manuals that are sent with the unit will NOT be returned.) To contact Fluke for information on how to have your units repaired, either:
- Visit www.fluke.com/37x-recall to register your affected product online as part of the recall. Once your information is complete and you click “submit,” a postage-paid mailing envelope will be sent to you within two business days. Use this envelope to return your clamp meter to Fluke for repair. Your clamp meter will be sent back to you when repairs are complete.
- Call 877-564-7180 for information and to request a postage-paid mailing envelope be sent to you.